ABB 266 pitot setup
I have a ABB 266 and I want to set up my device for flow measurement. How can I do that?.
Expert’s answer: ABB 266 pitot setup
The ABB 266 pitot setup is easy as ABC 123! But don’t get excited yet! You need info on the primary element to make all this work. Luckily for you, the ABB website or manual can give you all the info you to set up this meter. Still, it can’t hurt to go over it with you now.
To start with, the pitot generates dynamic pressure by creating a differential between its leading face and trailing face. Below you can see the equations to calculate the mass flow and volumetric flow.
Qm = K x √2 x Δp x density Qv=K x √2 x Δp x 1/density
- K = factor
- Density = density of the fluid
- Δp= differential pressure
- Qm= mass flow
- Qv= volumetric flow
You can set up the device easily through its display, as it gives you the relevant information step by step. You’ll need to set the lower and upper ranges of the process variable and choose the linearization type as well.
This gives you the basic configuration for volumetric flow. If you have a flow computer with temperature compensation and more, then the setup will take a little more work.
List of DP flow manufacturers:
Other resources that might interest you
Damping time and ABB pressure device setup
How d o you set up the damping time for an ABB pressure transmitter?
Differential pressure transmitter applied in a level measurement
Should I opt for a DP level transmitter if my level application is pressured and the product has stable density?
Supply isolator with auxiliary power for safe separation of 4...20 mA standard signal circuits