Expert’s answer: Changing flow units on Promag 400?
Changing flow units on Promag 400 with your display is pretty straightforward. First of all, you need the password to access the configuration, so make sure you have that. Once you use the password, you have access to the “Operator” and “Maintenance” sections. Each section offers specific information on the device.
After you access the menu setup, follow this procedure to choose the flow units:
- Setup menu > Advanced setup > System units
In the system units, you’ll see units for all your different variables, like volume, conductivity, temperature, mass flow, and more. Choose the one you want to change and you’re done!
Magmeter on the market:
Other resources that might interest you
What kind of information can I get from the Promag 400 with Heartbeat technology and will it make verification easier for me?
Promag 400 damping options!
Which damping option should you use for a flow meter?
Can you do a Promag 400 loop test with the display?
Can you use the display to do a loop test on a Promag 400?
Distance between sensor and transmitter – Promag 400
Is there a distance limitation in terms of how far a remote Promag 400 can be installed from a sensor?