Coriolis installation for gas application
Hey. We’re installing a new Coriolis in a gas application. Should we have inlet and outlet runs? What's the correct installation for gas?.
Expert’s answer: Proper Coriolis installation
Hey, so you want to know if a Coriolis installation requires inlet and outlet runs and how to correctly install it for a gas application.
Good questions. First, always read the manual before installing your device. Nobody likes reading them, I know, but it helps more than you think. Okay, now to your case.
The Coriolis meter doesn’t need inlet or outlet runs; this feature puts the Coriolis above the rest. You can install it vertically or horizontally, but when you mount it vertically, the fluid should go up. If you mount it horizontally and the temperature is high, install it head down. Otherwise put it in head up. Hope that helps!
Where can you get more information about Coriolis flow meters?
Want to know more about a Coriolis installation and where to find the right Coriolis flow meter? Is so, we’ve included some useful links below. Most of these companies have their own Coriolis flow meter installation guidelines that might be useful for you!
Companies with Coriolis flow meters:
Other resources that might interest you
Coriolis mass flow reading variation
What if there’s a variation in your Coriolis mass flow reading?
4-20 mA on Micro Motion 1700
What to do if the output isn’t working properly for a Micro Motion 1700 and you can’t access the device with the handheld?
Unreliable density reading
You have a CMF300 with 3500 RF Micro Motion Coriolis, but it’s not reading correctly. What can you do?
ROTAMASS Total Insight installation
How do you install the ROTAMASS Total Insight Coriolis from Yokogawa?