Differential pressure transmitter applied in a level measurement
2 minutes to read
I have a level application where I would like to implement a DP level transmitter, the tank is pressured and the product has a stable density, Is it the DP level transmitter the radar solution?.
Fabrício AndradeMar 06, 2017
Expert’s answer: Differential pressure transmitter applied in a level measurement
Before anything else, there’s something you should know: There are better technologies than a differential pressure transmitter on the market today that you can use for level measurement.
However, you need to figure out your budget for your application. That way, you can figure out if a differential pressure transmitter will provide you with enough information and if the accuracy is enough for your application.
Don’t get me wrong, I am not against applying a differential pressure transmitter in level measurement. Though I do believe that you have better options today. But for example, the density is a critical point when we talk about the level measurement using a differential pressure transmitter. If you haven’t changed the density values, this could be a good reason to use the DP transmitter.
The other point is the installation of the transmitter:
With the seal pot and the capillary, you can increase the level measurement’s risk for errors. That said, you have electronic differential pressure level transmitters today. When you use them, you don’t need to have seal pot or a capillary. So with this option, you avoid the old problems.
On the negative side, the electronic differential pressure transmitters are limited in terms of temperature. So depending on the application, you might not be able to implement this kind of technology.
If you still believe that differential pressure is the right solution for your application after reading these above-listed points, you have a list of DP pressure transmitter below. And, if you’re more interested in radars, I listed some of those, too!
In which kind of applications can I use a DP transmitter?
Rosemount 1151 temperature issues I read the Rosemount 1151 product data sheet and came across some temperature effect & temperature limitation issues. I would like to understand how temperature can affect my device and the measurement?
Radar or pressure transmitter? Nowadays, I see a lot of recommendations for how you can use radar for level measurement. Would it benefit me to switch from a pressure transmitter to a radar level meter?