Expert’s answer: Display information – PROMASS 100
Sure! It’s possible setup both the flow rate and also the totalization on the display in the Promass 100. However, you can’t configure the display settings within the display itself. The display is only for visualization. If you want to change the settings you will need a laptop or handheld.
The configuration is simple. The flow meter has a built-in web server with which you can do the configuration using a laptop without a specific software. You do not have a laptop or it is not possible to access the field with the laptop? No problem, you can make the setup the display with a handheld just as well.
Other resources that might interest you
Coriolis mass flow reading variation
What if there’s a variation in your Coriolis mass flow reading?
4-20 mA on Micro Motion 1700
What to do if the output isn’t working properly for a Micro Motion 1700 and you can’t access the device with the handheld?
Unreliable density reading
You have a CMF300 with 3500 RF Micro Motion Coriolis, but it’s not reading correctly. What can you do?
ROTAMASS Total Insight installation
How do you install the ROTAMASS Total Insight Coriolis from Yokogawa?