Emerson 5300 GWR and sludge!
We have an Emerson 5300 GWR LT! Sometimes, the level fluctuates severely. After we open and inspect the chamber, we find the rod clean but sludge in the chamber drain. After flushing the chamber, the level goes back to normal. How does this sludge cause this problem when the rod itself is clean? Can we use the GWR with sludge? .
Expert’s answer: Problem with Emerson 5300 GWR and sludge
Emerson 5300 GWR and sludge – jinkies, it’s a mystery, guys! Ruh-roh. Okay, now most folks who install level radars just need basic data about their applications. But depending on the process conditions and the tank, sometimes you gotta take it up a notch.
First things first, though. Everybody repeat after me: Check the manual.
So did you follow the installation instructions Emerson so kindly gave you? If you didn’t and the device won’t work properly, then you’ll probably find the reason in the mirror. Now review that manual and scratch that possibility off your list. While you’re checking the installation, scan for electromagnetic interference from a pump or motor nearby. You never know.
After you vet the setup, now you need to confirm the device configuration. For that, you need a handheld or a laptop with plant asset management software or the Rosemount Radar Master. When you hook up, check the basics and follow the setup guide. Someone may have tweaked the settings recently and that might mean your Emerson 5300 GWR and sludge don’t go too well together.
Once you’ve done that, you can check the echo curve analyzer for interference in the tank. If your mystery sludge meddled with your device, then the graphic should show the faults. If you find a ton of interference, you’ll need to do the surface threshold adjustment manually. Zoicks! You guys go ahead. We’ll be over here in the barrel, hiding.
Anyway, hope this helped you figure out what to do with your Emerson 5300 GWR and sludge!
No stories here, but if you want more on calibration, clicky clicky:
Other resources that might interest you
Output trim for Magnetrol Eclipse 705?
We need to zero trim our Magnetrol Eclipse 705 in the seal oil drain tank of a 7FA GE machine. What should we do – it asks for authorization when we connect it to the HART communicator?
Rosemount 3051S installation
Our Rosemount 3051S is exposed to temperatures around 100 degrees Celsius. We want to lower the temperature. How far away from the process can we move the device? Is there a limitation?
Endress+Hauser FMP51 output signal
The output for our Endress+Hauser Levelflex FMP51 isn’t changing proportionally with the level. Why do we experience this and what can we do?
Product review: Rosemount 5300 level transmitter
Want an unbiased review of Rosemount 5300? Then, read our product review.