Expert’s answer: Is there a flow meter with 0.05 percent accuracy?
Whoa! So, you need a flow meter with 0.05 percent accuracy? That kind of accuracy in flow meters is rare. Usually, flow meters have between 0.25 percent and 0.5 percent accuracy, and we consider these very accurate. There are only a few types of Coriolis that can meet your standards.
Of course, you need to scale out the flow meter for your process to check the accuracy in your full range.
So, check out Coriolis meters for a flow meter with 0.05 percent accuracy. These Coriolis meters offer just that:
If you want to scale out a flow meter online, here are online tools to help:
Other resources that might interest you
Calibration and Heartbeat Technology
If I have a report from the Heartbeat Technology, I don’t need to calibrate my flow meter, right? How can I prove that my device is calibrated and working properly?
Meter accuracy test
I have a magmeter measuring flow out of a tank. What’s the best way to perform a draw down test to check its accuracy?
Stainless steel turbine with mechanical counter!
I need a turbine flow meter with a stainless steel body and mechanical counter for a water application. Can anyone recommend a manufacturer?
PROFINET Flow meter
We have a PROFINET network in our company, and we are looking for a field device with the PROFINET already built-in. So, can we use the PROFINET connection?