There are three different types of Heartbeat technology. Let me explain to you the main differences of the three options:
- Heartbeat Diagnostic offers you a basic diagnosis of the flow meter which is based on the Namur NE 107.
- Heartbeat Verification provides you with a detailed device testing report. The report contains details about the output, the transmitter and the sensor. It also evaluates the current performance of your device. Furthermore, if you have FieldCare, you will gain additional details about the limits and a graphic of each verification, too.
- Heartbeat Monitoring does not only offer you the same information as the Heartbeat verification, it also enables an online monitoring of the measurement tube conditions.
For more details on flow meters you can check out these links:
You can read more about Endress+Hauser at Visaya Solutions:
Flow meter with HBT: First of all, the HBT is a proprietary technology for Endress+Hauser devices. The Heartbeat Technology does not offer you just one but three different types. Read more.
HBT Graphics: I want a graphic that depicts wear on my flow meter. This information could help in scheduling condition verifications to avoid unscheduled downtime. Can HBT do this? Read more.
FieldCare vs. Webserver HBT: The web server will give you a basic report that tells you what tests your device passed or failed using the Heartbeat technology. Most of the time, this report will suffice to tell you if your meter works properly. Read More.
Calibration and HBT: If I have a report from the Heartbeat, I don’t need to calibrate my flow meter? How I can prove that my device is calibrated and working properly? Read more