The difference between Heartbeat Technology Monitoring and Verification is easy to explain. With the Heartbeat verification, you either have to go to the device in the field or run the verification remotely through the web server or a PAM system. The Heartbeat Verification also provides you with a report covering all tested units of the flow meter.
The report from the web server tells you if the flow meter passed each step of the verification. The report from the PAM system provides the same information plus further measurement information, charts and graphics about previous verifications.
The Heartbeat monitoring transfers additional measured variables to an external condition system and helps you to detect modifications at an early stage.
Still not sure? You will find more details about the Heartbeat Technology in this video:
For more details on flow meters you can check out these links:
You can read more about Endress+Hauser at Visaya Solutions:
Flow meter with HBT: First of all, the HBT is a proprietary technology for Endress+Hauser devices. The Heartbeat Technology does not offer you just one but three different types. Read more.
HBT Graphics: I want a graphic that depicts wear on my flow meter. This information could help in scheduling condition verifications to avoid unscheduled downtime. Can HBT do this? Read more.
FieldCare vs. Webserver HBT: The web server will give you a basic report that tells you what tests your device passed or failed using the Heartbeat technology. Most of the time, this report will suffice to tell you if your meter works properly. Read More.
Calibration and HBT: If I have a report from the Heartbeat, I don’t need to calibrate my flow meter? How I can prove that my device is calibrated and working properly? Read more