The installation of the PROMASS 100!
We have a new device but no manual. Could you tell us how to correctly install the Promass 100? .
Okay, here we go again! The Promass 100 follows universal Coriolis rules. That means it has the same installation basics as other flow meters on the market.
Here are the rules of the Coriolis club:
- Install it in a vertical pipe with the flow going up whenever possible.
- If you have to install it horizontally and it has bubbles or gas, head up again.
- If you have to install it horizontally and it has solids in the fluid, head down this time.
- Never install it with the head to the side.
- It doesn’t need a straight pipe before or after it.
- If you have a pressure transmitter after it, you need three to five outlet runs.
- If you have a temperature transmitter after it, you need four to eight outlet runs.
Look up more flow meters here:
You can read more about PROMASS 100 at Visaya Solution:
EtherNet/IP PROMASS: Integrating an EtherNET/IP device into the Rockwell system is easy. The PROMASS 100 offers a seamless integration with different types of control systems. Read More
PROMASS into EtherNet/IP: The Promag 100 should have Dynamic Host Configuration Protocol (DHCP). That means that when you connect the device to the EtherNet/IP network, the flow meter will assume an IP address. Read more
PROFINET Flow meter: To be honest, there are already better options on the market nowadays. The Emerson Automation Solutions offer the Micro Motion 5700 with PROFINET, EtherNet/IP and Modbus. Read More