Yes, of course, we can show you how to check the K-factor for Rosemount 8712. This one is very easy. On the interface of the transmitter, you should find a button labeled “Tube Cal No.” Press that and, voila! It will show you the K-factor right away.
Now, if you want to access the configuration using a handheld, then use the following steps:
Device > Basic Setup > Calibration Number.
That should give you exactly the same figure as the Tube Cal No. button.
If you want to read more about the 8712 setup, click here!
Magmeter on the market:
Other resources that might interest you
Rosemount 8712 damping time
What’s the maximum Rosemount 8712 damping time?