Micro Motion 1700 output
Help! I tried to set up the 1700, but the output isn't working properly and I can't access the device with the handheld. We managed the basic configuration from the front display. Locally the device works fine, but the output doesn't reach my control system, which means the process values don't either. What do I do?.
Expert’s answer: Micro Motion 1700 output
Hey, we got this! The Micro Motion 1700 output is an active signal, so you don’t need a power supply. Pay attention to your loop installation – maximum loop resistance cannot exceed 820 ohms.
If the values are within the correct intervals, then do a quick check using a loop tester. Simulate a 4-20 mA output and inspect your cables to cross off structural problems. A faulty cable could be the culprit.
Now grab the handheld and dive into the HART interface. (If you need help connecting the handheld, see the pictures below.)
Before connecting the signal cables to the Micro Motion 1700, use a multimeter to confirm you have voltage from your device. Connect the signal cables and simulate the output signal using the built-in function:
Simulate > Simulate Outputs > mA output loop test (select the mA)
Now you can do the configuration. If you’ve checked all of the above and still have trouble, you may need to contact Emerson’s tech support, as the problem could be with the device itself.
Configuration and use manual: Micro Motion® Model 1700 Transmitters with Analog Outputs
Installation manual: Micro Motion® Model 1700 and 2700
More companies with Coriolis flow meters:
Other resources that might interest you
The best flow meter: What you need to know
I’m looking for the best flow meter for my application. What’s the best magmeter for me?
TOP 5 Coriolis flow meters on the market
Want to know which Coriolis flow meters make it to our top 5 list? Click to find out!
Non-intrusive flow meter
What rubbers should we choose for a vortex meter that’s used to meter his gross production in a heavy oil field (API 13)? Is there another option than a Liquid Ultrasonic or Cidra Sonartrac?