Output trim for Magnetrol Eclipse 705?
Hey, we have a Magnetrol Eclipse 705 in the seal oil drain tank of a 7FA GE machine. I need to zero trim it, but when I connect it to the HART communicator, it asks for authorization. Help?.
Expert’s answer: Output trim for Magnetrol Eclipse 705
Need help with output trim for Magnetrol Eclipse 705? Okay, it sounds like someone fooled around with your setup. Hence, I took a look at the 705 manual. Lo and behold, I found a note about the password for configuration access. See for yourself:
The default user password … is 0. The last step in the configuration menu provides the option to enter a new password. With a password of 0, the transmitter is no longer password protected, and any value in the menu can be adjusted without entering a confirming password, except diagnostic values.
So someone may have changed the factory password. But, if you don’t know it, Magnetrol recommends contacting local support. For more details, check out the manual.
Where can you get more information about output trim for Magnetrol Eclipse 705?
Want more information on output trim for Magnetrol Eclipse 705? Here are some links that might help you out.
You can see the market’s current level solutions here:
Other resources that might interest you
Level transmitter FMR51 error
Why does the radar for FMR51 read correctly, except for when the mixer is on and the level is below the propellers? When this happens the signal reads full scale.
Level transmitter calibration!
Do you need to calibrate level transmitters?
Level transmitter malfunction
An Endress+Hauser Levelflex FMP51 is installed on a diesel tank for level measurement, but its output is not changing proportionally to the tank level. Why?
DP transmitter level calculation
What information do you need to select the pressure cell for a differential pressure transmitter to do level measurement?