Expert’s answer: How to connect a pressure transmitter and HART communicator
Hi! Connecting a pressure transmitter and HART communicator? Then, let’s do this!
Transmitter connection – Pressure transmitters usually use current loops for power. That means you have 24 volts of direct current (VDC) in the same loop where you have 4-20 mA and HART. Now, to send data with the device in the workshop, you need a 24-VDC power supply, a 250-ohm resistor, and a handheld. Look at the diagram below to see how to install the device and connect the handheld.
In the field, you shouldn’t need the resistor because your cable resistance should suffice. But if it’s not enough for some reason, just use the same setup you had in the workshop.
For more information about HART, please go to www. hartcomm.org.
For more information, please check with these companies:
Other resources that might interest you
Rosemount 3051S installation
If we have a Rosemount 3051S that’s exposed to an air temperature of 400 degrees Celsius and 3/8-inch stainless steel tubing, how do we lower that number? How far out can we go?
Pressure transmitter basics
What are the different types of pressure sensors available in the market? Here’s what you should know!
Pressure transmitter level calculation
Which type of pressure transmitter should I use for level measurement if I have a 10-meter tank open to atmosphere with a product density of 0.859 ρ.? How do I calculate the level range?