Promass 200 installation!
Is there a procedure to install the Promass F 200? I need it for a liquid application, but I don't have the manual. Could you help?.
Expert’s answer: Promass 200 installation
Yup, up! No, that’s not a typo. Let me explain. No, there is too much. Let me sum up what you need to know about doing a Promass 200 installation.
Coriolis meters have pretty easy setups compared to most other meters. However, you have to follow one simple but super-important rule to get good results. Up. Up up up.
Install your Promass F 200 in a vertical pipe with the flow going up. That way, you know the pipe will always exist in one of two conditions, filled or not filled. Partially filled pipes won’t cut it. Installing it this way will avoid a lot of future problems. If your operator likes to say that the meter doesn’t work properly, then this setup should kill that excuse dead.
If you can’t install the flow meter in a vertical pipe, you’re stuck. Just kidding. You have other options, but they’ll depend on the product you need to measure. Just go up. You’ll thank me for it.
Look up more flow meters here:
Other resources that might interest you:
PROMASS 200 and Flow Meters Go Wireless
Can you use a PROMASS 200 in a wireless HART network? Here’s what you need to know!
Measuring natural gas with Promass 200
We need a loop-powered device to measure natural gas flow. Are there devices that are accurate enough?
Multifunctional field indicator for monitoring and displaying analog measured values
Multifunctional switch panel process measuring device for monitoring and displaying analog measured values