Promass 200 with PROFIBUS PA
Can I have a loop-powered Promass 200 in a PROFIBUS network? I need to know about the power consumption of the flow meter..
Expert’s answer: Promass 200 with PROFIBUS PA
You certainly can pair a Promass 200 with PROFIBUS PA. The Promass 200 offers the loop-power option to the HART, FOUNDATION Fieldbus and PROFIBUS protocols, so you don’t need a separate power supply for it.
When we talk about the PROFIBUS network, it helps to know the device current output and consumption. The voltage for the PROMASS 200 ranges from 9 to 32 volts of direct current. If you have only one output working, then you should have a current of 16 milliamps and consumption of 1:512 milliwatts. You’ll get 2:2512 milliwatts when you have two outputs in operation.
Coriolis 2 wires on the market:
Other resources that might interest you
PROMASS 200 and Flow Meters Go Wireless
Can you use the PROMASS 200 in a wireless HART network? And can the flow meter be powered by a wireless adapter?
PROMASS 200 installation
How do you install the Promass F 200 for a liquid application?
Natural Gas – Promass 200
We need a loop-powered device to measure natural gas flow. We want the Promass F 200 or the Micro Motion, but we need to know if we can use these devices in our system and if they’re accurate enough?