So, you want to know how to find the lower and upper range values of a differential pressure transmitter? If you know the range of the flow transmitter and the maximum flow, you can calculate any flow value. Take a look.
Maximum flow: Max flow = K x √ΔPmax
Constant: K = max flow/√ΔPmax
Differential pressure: ΔP = (flow/K)²
Output (4-20 mA): Output mA = [(%ΔP) x 16 mA] + 4 mA
The orifice plate should have some information that can help. You can also contact the company that made the orifice plate for more details. These calculations give you the simplest ways to find the flow, ?P or the output. Your operator can provide the process values and maximum flow if you have trouble finding this data.
Where can you get more information about lower and upper range values of differential pressure transmitter?
Want more information on DP flow manufacturers? Here are some links that will get your started!
List of DP flow manufacturers:
Other resources that might interest you
DP transmitter level calculation
What information do you need to select the pressure cell for a differential pressure transmitter to do level measurement?
Differential pressure applications
In which kind of applications can you use a DP transmitter?
Pressure transmitter data
How do you check if a pressure transmitter S/N K219C601052 from Endress+Hauser is providing the required data?
Static and dynamic pressure
What’s the difference between static and dynamic pressure?