Rosemount 8712 damping time
What's the maximum damping time I can set up for the Rosemount 8712?.
Expert’s answer: What’s the Rosemount 8712 damping time?
You’re right, there’s a maximum Rosemount 8712 damping time. The device has its damping range limited to between 0.0 and 256 seconds. However, you need a good setup to avoid problems in the process control. Some technicians change the configuration, which winds up creating a bad value output.
Unfortunately, this change won’t solve a measurement problem. If your device setup hides the variation, then you ruin your measurement accuracy.
You should only use the damping configuration to stabilize small shifts in the process measurement. If you have a larger variation, then you should try to figure out the problem rather than raise your damping to obscure it.
Magmeter on the market:
Other resources that might interest you
Rosemount 8712 testing
Is there a tool that can simulate a flow meter?
Compressed air flow measurement on Rosemount 8800D
What to do when you experience a problem with compressed air flow measurement with the Rosemount 8800D vortex flow meter?
Multifunctional switch panel process measuring device for monitoring and displaying analog measured values