Unreliable density reading
We have a Micro Motion Coriolis, the CMF300 with 3500 RFT, to measure the density of condensate in a custody metering skid. Now it's not reading correctly. It shows almost 1000 kilograms per square centimeter, but it should be around 700. How do we solve this problem?.
Expert’s answer: Unreliable density reading
So, you’re having an unreliable density reading? Interesting. Let’s verify your setup. If the meter still reads incorrectly afterwards, you should contact support at Emerson.
Okay, to troubleshoot this issue, you need to eliminate errors that could cause or contribute to it. First check the installation, using the CMF300 installation manual or the CMF300 user manual. Make sure you apply the standards for your particular setup.
Next, check for kinks in your system from scale out. If you installed the meter a year ago, then put in another device three weeks ago, the new device may have caused the problem. So always make sure your devices can support or interact with new devices.
Finally, try using a diagnostic tool on the CMF300 to check your setup and show error messages. Here, the Micro Motion has software to test the device conditions, Prolink. Depending on your device, you may have the meter verification function, giving you data on the flow meter conditions.
For more information, check out the CMF300 installation manual or user manual. There you should find notes related to density measurement such as these:
- Process temperature effect is the change in density accuracy due to temperature change away from the calibration density.
- Process pressure effect is the change in flow and density accuracy due to pressure change away from the calibration pressure. This effect can be corrected by dynamic pressure input or a fixed meter factor.
Want to learn more about how to solve an unreliable density reading?
More companies with Coriolis flow meters:
Other resources that might interest you
Yokogawa ROTAMASS flow direction
Last week our ethylene flow meter suddenly went to zero in the DCS, but the actual flow was normal. We reversed the flow direction, and the flow became normal. Now the actual flow direction is forward, but the transmitter configuration is reversed. What’s going on?
Best method to calibrate displacement sensors?
What’s the best method to calibrate displacement sensors?
Coriolis mass flow reading variation
We have three Coriolis mass flow meters for ethylene, one on the main battery limit and the other two on different reactors. Their meters should match the main battery’s meter. Right now, we have one to two tons per kilogram of difference, with the battery limit showing less. Can anybody help me solve this issue?
PROFINET Flow meter
We have a PROFINET network in our company and we are looking for a field device with the PROFINET already built-in. How about the PROFINET connection?