Expert’s answer: Upstream and downstream pipes for Rosemount 8712?
Do you need to install upstream and downstream pipes for Rosemount 8712, you ask. Then you need to know that nearly all brands have the same installation specs. Also, you should have little trouble remembering them, as they’re fairly easy. Ready?
Okay, before the device you need five straight pipe diameters upstream. Then after the device, you need two pipe diameters downstream from the electrode plane. There you go!
If you install the device using these simple rules, then you should get the accuracy stated in the manual.
Want more information? If so, these companies have magmeters on the market:
Other resources that might interest you
Magmeter for slurry!
We plan to buy a magmeter. Which liner works best for slurry? Here’s what you need to know!
Best magmeter for water application
Want to know which type of magmeter is best for a water process flow measurement? Then, read on to find out!
Effect of electromagnetic pipeline mapper on magmeter?
Can an electromagnetic pipeline mapper produce any specific effects on a magnetic flow meter? Click through to find out!