VEGAPULS 64 setup
We have a new VEGAPULS 64 for our level application! But I have doubts about how to set up the transmitter. Could you explain?.
Expert’s answer: VEGAPULS 64 setup
No problem! You can do the VEGAPULS 64 setup even before you install it, which may make things easier. As soon as you turn on your device, you can configure it through the display, Bluetooth, or your plant asset management system.
I recommend using the Quick setup option. It’ll give you a step by step guide through the VEGAPULS 64 basic configuration. However, if you want to customize your setup, then you can use the standard option. There you set everything manually, like your measurement loop name, units, application, and adjustment.
The VEGAPULS 64 allows Bluetooth communication as well. Just download the Vegatools app and set up the device with it, following the same concept.
Other resources that might interest you
What to do with VEGAPULS 62 setup error message?
My VEGAPULS 62 is showing an error message. What to do?
Radar interference – VEGAPULS 61
What to do when a VEGAPULS 61 shows the wrong level?
VEGAPULS 64 GSD file!
Where can you find the GSD file for the VEGAPULS 64?