Expert’s answer: The basics of a Yokogawa Rotamass setup
I suppose that you have a Rotamass 3 and not a Rotamass Total Insight? In that case, a Yokogawa Rotamass setup is not at all complicated. You can do the setup directly on the display. Alternatively, if you want, you can do it through the HART or whichever protocol that you have.
First of all, you need to access the main menu through the display. Once there, roll down the options up to “Display configuration”. This is the option that you’re going to use to do the setup. To show the process variable on the display, you have the options to select 1, 2 and 3.
The variable on the display should be the information most important for the operator and for you when you need to go to the field to check the process information. As soon as you have a digital communication, all process information can be accessed in the control system. If you’re only using an analog communication the 4-20mA will be related to a unique process variable.
Want more information? If so, below are more Coriolis flow meter options on the market:
Other resources that might interest you
TOP 5 Coriolis flow meters on the market
Do you have questions about which flow meter you should choose for your application? Here’s what you need to know!
Yokogawa ROTAMASS flow direction
Do you have any experience with ROTAMASS? Last week our ethylene flow meter suddenly went to zero in the DCS, but the actual flow was normal. We reversed the flow direction, and the flow became normal. Now the actual flow direction is forward, but the transmitter configuration is reversed. What’s going on
ROTAMASS 3 loop test
Could you teach me how to do loop testing with the ROTAMASS 3? Can I do it using the display?