We need to set up a pressure transmitter on an orifice plate for flow measurement. How can I calculate the flow range to set up the transmitter?
Expert’s answer: Flow range for orifice plate
Usually when you set up a differential pressure transmitter, you need to decide what variable the device will measure, such as flow, level, volume, mass, and others. Then you can use the equations below to find your flow range for orifice plate.
Maximum flow: Max flow = K x √ΔPmax
Constant: K = max flow/√ΔPmax
Differential pressure: ΔP = (flow/K)²
Output (4-20 mA): Output mA = [(%ΔP) x 16 mA] + 4 mA
List of differential pressure flow manufacturers:
Other resources that might interest you
Vortex flow meters in low flow profile
I never knew the impact vortex flow meters could have until I saw my signal trends disappear in low flow mode. So what applications suit vortex meters best?
Differential pressure applications
In which kind of applications can I use a DP transmitter?
Lower and upper range values of differential pressure transmitter
On my differential pressure transmitter, I have a range from 0 to 500 psi. We use an orifice plate for creating differential pressure. Maximum DP in our gas flow is about 200 psi but it’s not fixed. Can you tell me how to determine the lower and upper range values?
How will a vortex flow meter in my steam flow measurement benefit me?