HART pressure transmitter: How to set up two displays
Hello. I have a HART pressure transmitter in the field, and I want a setup for two displays in different locations for the same pressure device. Can I do that? I’m using only the 4 – 20 mA.
Expert’s answer: Can you set up a HART pressure transmitter for two displays?
You want to know how you can set up two displays in different locations for the same HART pressure transmitter… A quickie! My favorite. Short answer: YES. Long answer: Even if you only have the 4 – 20 mA, if your device has HART, you can do it! You definitely need a HART display. You’ll also need a physical network.
Once you have these two goodies, you can set up two masters in the HART communication. Never mind that old saying about two masters. We ain’t talking philosophy here! We’re talking science! You want two, you get two! Back to business. Next, configure one display as master 1 and the other display as master 2. Both should read the same HART values from your field device. If they don’t, you did something wrong. Check your setup and try again.
Okay, now that you’re set up and getting good data, pick your values. You can have four, conveniently named primary, secondary, tertiary, and quaternary. Just configure whatever you need to show in the displays. And you don’t even have to do multidrop installations, as the displays will work as Handhelds. Boom!
Want to learn more and maybe find a suitable HART pressure transmitter? Check these links that we’ve put together.
The market has many such options. Here are a few:
Data acquisition on the market:
Other resources that might interest you
Absolute, gauge, and differential pressure
Should you use a differential pressure transmitter or a gauge or absolute pressure transmitter to measure the pressure in a water pipe?
Rosemount 3051S installation
A Rosemount 3051S pressure transmitter needs to be moved away from the process. But how far out can you go?
How do you connect a pressure transmitter with a HART communicator?
Offset and slope for differential pressure transmitter?
How do you do the offset and slope adjustment when installing and calibrating a Rosemount differential pressure transmitter?