Hey, why do electromagnetic flow meter have different liners?
Expert’s answer: Why are there different magmeter liners?
Take a moment to think about it. The magmeter liners always come in contact with the products. And some products react to those liner substances. Therefore you need to choose your liner carefully, unless you have a fairly inert product like water.
For example, if you have a corrosive product, you need a liner material that has a chemical compatibility with the product. Teflon makes for a good choice, as it gives you a wide range of product applications. If you see options like polytetrafluoroethylene (PTFE) and perfluoroalkoxy alkane (PFA), then you should know that they mainly differ in temperature limits.
On the other hand, for abrasive products, you have a variety of rubber and ceramic options you can consider. But that depends on your budget.
So there you have it. Nearly every brand you check out will have most of these options available for their magmeters. Just remember your product’s chemical compatibility!
Magmeters on the market:
Other resources that might interest you
Magmeter for slurry!
We plan to buy a magmeter, so which liner works best for slurry?
Best magmeter for water application
Which type of magmeter is best for a water process flow measurement?
Effect of electromagnetic pipeline mapper on magmeter?
What effects can an electromagnetic pipeline mapper produce on a magnetic flow meter?