What's the standard flow accuracy of PROMASS 100? If I want to have a better performance is it possible with that device?
Expert’s answer: Mass flow accuracy of PROMASS 100
The stated mass flow accuracy of PROMASS 100 is +- 0.35% for gas mass flow. For liquid mass and volume flow it is +- 0.10%. However, it is possible to get an even better performance if you want. This option is called PremiumCal for mass flow and the accuracy will be +-0.05%.
Although, the accuracy will depend on the process information and you can’t have a variant of the flow meter accuracy. The right way to know the exact flow meter accuracy is simulating the process conditions on the application tools and then scale out the right flow meter for your application.
The software will show you the precise accuracy throughout your process conditions so you can check if the accuracy that is stated in the technical manual will stay the same.
Other resources that might interest you
Coriolis mass flow reading variation
What if there’s a variation in your Coriolis mass flow reading?
4-20 mA on Micro Motion 1700
What to do if the output isn’t working properly for a Micro Motion 1700 and you can’t access the device with the handheld?
Unreliable density reading
You have a CMF300 with 3500 RF Micro Motion Coriolis, but it’s not reading correctly. What can you do?
ROTAMASS Total Insight installation
How do you install the ROTAMASS Total Insight Coriolis from Yokogawa?