Measuring natural gas with Promass 200
My company needs a loop-powered device to measure natural gas flow, but we need to know if we can use these devices in our system and if they’re accurate enough. Could you help us?
Expert’s answer: Measuring natural gas with Promass 200
Either of these devices would work for your application. However, the Micro Motion doesn’t have a real 4-20mA output. The meter has a barrier to rescale the 12-20mA from the field to 4-20mA for the programmable logic controller (PLC). That’s why measuring natural gas with Promass 200 might be the right thing for you.
You see, the Promass 200 offers a real 4-20mA output with the modulation in the whole range, so you don’t need a barrier to convert.
Both devices will provide good accuracy for your application, and best of all, you can simulate the performance with online tools from both brands! Try before you buy!
Want more information? If so, check these online tools:
Two-wire Coriolis meters on the market:
Other resources that might interest you:
PROMASS 200 and Flow Meters Go Wireless
Can you have the PROMASS 200 in a wireless HART network? And can the flow meter be powered by wireless adapter?
Promass 200 with PROFIBUS PA
Can you use a loop-powered Promass 200 in a PROFIBUS network? Read more.
Promass 200 installation
How do you install a Promass F 200 for a liquid application? Here’s what you need to know!