After 10 years, we want to change our flow meter to a new one. Could you inform us if the PROMASS 100 will work for our application? The process information is: 0 - 100 kg/min, Milk, ambient temperature and 2 bars a of pressure. Thanks.
Expert’s answer: Promass 100
Hi there, sure I can.
Wow, 10 years with one flow meter means a lot of patience! Your intention is totally understandable. I would say, the easiest way to check if the flow meter will match your application is using the online tool and simulate the process conditions. Regarding the Endress+Hauser device, the online tool means the applicator.
The applicator tool shows that a PROMASS 100 sized 1” will provide you with accuracy around 0.1% and a pressure loss of 0.39 bar. If you would like to check the same process conditions for other brands, you can find similar tools online.
If you want to scale out a flow meter online, here are online tools for you:
Hope you will find a perfect flow meter match!
Other resources that might interest you
Coriolis mass flow reading variation
What if there’s a variation in your Coriolis mass flow reading?
4-20 mA on Micro Motion 1700
What to do if the output isn’t working properly for a Micro Motion 1700 and you can’t access the device with the handheld?
Unreliable density reading
You have a CMF300 with 3500 RF Micro Motion Coriolis, but it’s not reading correctly. What can you do?
ROTAMASS Total Insight installation
How do you install the ROTAMASS Total Insight Coriolis from Yokogawa?