Can we connect two temperature transducers to one RTD? We need to maintain the temperature gauge and somehow place a transducer device to get a 4-20 mA output. Do we need to disable the existing gauge and replace it with a temperature gauge with this output?
Expert’s answer: How to connect one temperature sensor and two transmitters?
So, you want to connect one temperature sensor and two transmitters. Unfortunately, adding a transducer to your resistance temperature detector will reduce the correct resistance for both transmitters. If your current transducer provides 4 to 20 milliamps, you can install a process indicator remotely to check the values. Better yet, install a transducer with HART communication and remote display for the values and device status.
The installation of the HART display will work as a secondary master. If you need a field communicator, you’ll usually have a handheld as a primary master. Also, the display can be a primary master, depending on your need.
The market has many such options. Here are a few:
Wireless solutions can help here, because you won’t need to invest in cables and structure to receive your data. As soon as you have the field device and the gateway set up, your control system can easily pull the values. Wireless also receives the diagnostics from your field device, and you can even set up the device remotely.
Hope this answered your question on connecting one temperature sensor and two transmitters!
Other resources that might interest you
Wireless temperature measurement and logging
We have an application where requires a temperature measurement and logging, on up to 24 separate products for 12 hours. We always have cables issues and I would like to know if our application can be measured by a wireless solution, could you clarify?
Temperature loop transmitter?
I have a temperature loop system where I am receiving all information from the field directly into the PLC without a transmitter, It means that PLC is reading the RTD and thermocouple directly. Is there any advantage to apply a transmitter in the field?
Temperature measurement between two parts
In reality, I could likely settle for thermocouples one/inch or so, but that would be a lot of inches at 36×36″. Does anyone know any devices, sensor matrix, and or material that may work here?
Temperature transmitter calculations
Our temperature transmitter has an input range of 0 to 100 degrees Celsius, an output range of 4-20 milliamps, and accuracy of ±0.5 percent. So, if the output of the transmitter is 13.6, what are the ideal minimum and maximum temperatures?