I need to change our cutoff value because the operator says the magmeter is measuring without flow. How can I change the low flow cutoff on a Rosemount magnetic flow meter 8712?
Rosemount magnetic flow meter 8712
Let’s see. The Rosemount magnetic flow meter 8712 has a range between 0.01 and 38.37 feet per second, and yes, you can change the cutoff value. Very good.
Okay, to adjust the cutoff in the Rosemount magnetic flow meter 8712, just follow these steps:
Device > Detailed Setup > Signal Processing > Low Flow Cutoff
Then input the values you want. Keep in mind though that you need to set up the flow meter with the values between 0.01 and 38.37 feet per second. Below that, the flow will be zero.
If you have a handheld from Emerson, you’ll do the step by step on the keypad:
1, 4, 4, 4
Hope that helps!
Magmeter on the market:
You can read more about magmeters:
Magmeter or DP meter in energy monitoring system industrial – We’re implementing an energy monitoring system, and we can use either a DP wireless or a magmeter. What are the differences between these solutions? Read more
Magmeter connection, Promag P 500 – I have the new Promag P 500 and would like to know which cable and connection should I use to connect the sensor to the remote transmitter? Read more
Magmeter liners – Take a moment to think about it. The liners of magmeters always come in contact with the products, and some products react to those liner substances. Therefore you need to choose your liner carefully, unless you have a fairly inert product like water. Read more