I want to test a thermocouple loop by injecting a millivoltage from the field where the thermocouple terminates in its compensating cable back to a substation and a thermocouple/mA converter card with cold junction compensation. Let's say the field is 30 degrees Celsius and the substation is 10 degrees Celsius.
Expert’s answer: How to use thermocouple basics to test a thermocouple loop
Piece of cake! This goes all the way back to thermocouple basics. Come, children. Let’s have our lesson for the day.
Now, you’ll need to sum up the millivolts (mV) for each temp that you have from the sensor to the input card. The temp where you have the card will also come into play here, so get that too.
When you’ve corralled all these numbers, you’ll find that your error margin is minimal most of the time. Take a look at this handy-dandy pic to give you a better idea of what you need to do!
These guys have temp transmitters:
Check the Fluke 52 II Digital Thermometer on Amazon U.S. and Germany:
Other resources that might interest you
Can thermistors detect temperature ranges up to 1600 degrees Celsius? What device can handle regular temperatures between 1100 and 1200 degrees Celsius, with excursions up to 1300 or 1400?
Is there an industrial-rated handheld that accepts TC type J, K, T, or two- or three-wire RTD inputs with either a corresponding mv output or 4-20 mA output?
What are some temperature measurement questions you MUST know? Find out here!