I never knew the impact vortex flow meters could have until I saw my signal trends disappear in low flow mode. So what applications suit vortex meters best?
Expert’s answer: What applications suit vortex flow meters in low flow profile?
So, you want to know what applications suit vortex flow meters in low flow profile? All flow meters have limitations. It’s important to understand your process and the accuracy required for it before selecting a flow meter. Vortex flow meters have a broad range of applications in gases and liquids, but they all need a minimum velocity to begin measurement. Once your Reynolds number goes below 20,000 (15,000 for gases and steam), the K-factor becomes non-linear. The non-linearity depends on the process fluid’s density and viscosity.
Usually, the manufacturer will send a simulation report, explaining the behavior of the flow meter in the process condition. So when you or your engineers purchased a vortex for your process, you should have received some sort of data regarding the limitation on the flow measurement.
You may need to add another flow meter to the process for the low profile measurement. This kind of implementation is typical when you have a low profile and you need the full range measurement. A thermal flow meter should take care of the low profile, leaving the vortex to measure the flow.
Want to know more about vortex flow meter applications?
Here, check out our recommended links for flow meters:
Other resources that might interest you
Online diagnostic of vortex flow meter?
Is there any way to check your vortex flow meter online?
Reliable steam measurement?
What’s the best technology to measure high-pressure steam flow? Also, is there a portable instrument for measuring steam?
Is a vortex a good solution for measuring your air flow? What limitations does it have?
Orifice plate or vortex for steam flow meter?
Which do we recommend – a vortex flow meter or an orifice plate?