WirelessHART with Yokogawa AXR
We have the Endress+Hauser wireless adapter SWA70, and we intend to use a Yokogawa magmeter. Which kind of PV, SV, TV, and QV information should we get from the HART?.
Expert’s answer: WirelessHART with Yokogawa AXR
Hey, good question! In short, you can get the required information if you use a WirelessHART with Yokogawa AXR. So that means the SWA70 will read the HART values from the Yokogawa device. But depending on the process values, you may need to change the HART values to read remotely.
Anyway, you should get these HART standard values from the Yokogawa AXR:
- PV – Instantaneous flow rate
- SV – Forward totalization value
- TV – Reverse totalization value
- QV – Differential totalization value
If you want to learn more, then check out the manual!
Alternatively, check out these companies with magmeters on the market:
Other resources that might interest you
Yokogawa AXR loop test
Can I do a loop tests directly on the display with field flow meters from Yokogawa? That’s exactly what we look at in this article.
Yokogawa AXR electrodes
Do magmeters with more than three electrodes differ somehow from the Yokogawa AXR? Click through to find out!
WirelessHART configuration basics
Want to get to know the basic configurations for a new wirelessHART network? Then, here’s what you need to know!
Want to know whether it’s possible to send digital communications through analog signals? If so, this is what you need to know!